Ensure this is done before starting. I agree, user guide is written swell, what gets my goat is board’s defiance. It’s a bit frustrating to have all logic designed and simulated and no way to see if it really works. Was this Answer Record helpful? We have detected your current browser version is not the latest one.
|Date Added:||5 November 2016|
|File Size:||34.90 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Try it on a different computer. As for broken USB – I’m not sure what’s wrong with it. You can safely assume that the clock is toggling from high to low and back again at every sample point.
Description This answer record provides an overview of how to manually install Digilent Programming Cable drivers.
Solved: Digilent JTAG USB Cable and chipscope – Community Forums
These instructions assume the cable driver installation has been selected when installing Xilinx tools and that the driver files have been extracted to the machine by the installer. Have you verified that all of the connections are correct between the Digilent cable and the board? The Digilent plug-in software and dgilent drivers must be installed on your machine for you to be able to run the applications from the network.
We have detected your current browser version is not the latest one. To make libusb 1.
Was this Answer Record helpful? Ensure this is done before starting. So yo have lots of optionsall that have to be in the correct way for programming to work.
Solved: Digilent JTAG USB Cable – can’t initialize chain – Community Forums
No this is not right. If the above procedure does not help, try uninstalling the Digilent driver and Xilinx tools and start fresh with a new Xilinx tools install. Adrpt I mutilate the cable ending and get rid of this gray ‘thing’?
Check that the cable, scan chain, and power connections are intact, Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. ChromeFirefoxInternet Explorer 11Safari.
Having all ones returned is an indication that the chain isn’t correct. I tried with platform cable II and flash is being programmed. Here’s what Adept spits out on Win7: I’m not sure the Xilinx Tool plug-in currently posted on the Digilent website is compatible with HS1.
Description Solution Linked Answer Records. So what do you suggest?
My boss gave me the board, gave me the cable and said ‘use this, do not use USB’. You can’t sample your system clock if you’re clocking the ChipScope core with the same system clock. Although jtag scan cannot find them but the last program which digilfnt downloaded on PROM when everything was okstill works.
ChipScope core is triggered at the positive edge of the system clock As requested, ChipScope samples the signal of interest, which happens to be the system clock, and finds it high see previous point Clock signal in waveform display appears to always be high You can safely assume that the uusb is toggling from high to low and back again at every sample point.
Thanks for all the replies, I really appreciate it. You need to read the User Manual for this board which explains the configuration option. adetp
If the above procedure does not help, try uninstalling the Digilent drivers completely. We have detected your cigilent browser version is not the latest one.
The Digilent website indicates that there is a V2. It’s out withthe scope, to see what th esignals on the JTAG chain look like when your programming. If not you should before posting.